## Antenna - Theory .com: The Antenna Site

antenna-theory.com :: View topic - Dipole antenna characterization
antenna-theory.com
Antenna Theory

Author Message
Mohican
Antenna-Theory.com Newbie

Joined: 09 Mar 2015
Posts: 2

 Posted: Tue Mar 10, 2015 4:07 pm    Post subject: Dipole antenna characterization Hi Guys, I have a question about dipole antenna characterization with VNA. Generally, VNA has signal and ground in its output port. Let's assume that left part of dipole (connected to core and gets signal from VNA) so it has regular quarter wavelength current distribution. Then, the right side of dipole, that suppose to be specular reflection of left one, has constant current distribution=0, since it connected to coating, that is grounded. 1. So since half of dipole antenna has a constant potential (viz. ground), does it mean that only half of antenna radiates,in other words dipole act as a monopole? 2. Does the fact that half of the dipole is grounded affect the measured impedance of the antenna? Regards, Newbie

Joined: 03 Jan 2007
Posts: 197

 Posted: Tue Mar 10, 2015 7:53 pm    Post subject: Ground does not have zero current through it ground is a path for rf current to flow
R. Fry
Antenna Theory Regular

Joined: 06 Jun 2011
Posts: 42
Location: Illinois USA

 Posted: Wed Mar 11, 2015 10:33 am    Post subject: The inner surface of the outer conductor of a coaxial cable carries the same r-f current magnitude as the outer surface of the inner conductor, just 180 degrees out of phase with it. When a coaxial cable is connected to the feedpoint of a dipole antenna, r-f current flows out of one side of the dipole and into the other side, changing directions during each cycle of the r-f waveform.
helix
Antenna Theory Regular

Joined: 29 Jan 2015
Posts: 64

 Posted: Mon Mar 16, 2015 12:25 am    Post subject: The VNA coaxial output is "unbalanced" (asymmetric w.r.t. ground) while the dipole structure is "balanced" (symmetric w.r.t. ground). You will want to use a device called a balun (balanced to unbalanced converter) to measure the dipole input impedance correctly with VNA. The VNA is connected to the unbalanced port and the dipole arms are connected to the two balanced terminals). If you do not use a balun, then in general you will find that the VNA cable (which you generally don't intend to be part of your antenna/radiating structure) will be "hot" (have "antenna mode" RF currents on it) and this will affect your VNA impedance measurement to various degrees, depending on the circumstances. A basic test to see how "hot" the VNA cable is is to run your hand along its length and see if the antenna impedance (i.e., Smith chart locus) changes. Ferrite (absorber) beads clamped to the VNA cable can help suppress this effect, but the real problem is that the antenna is not fed in a balanced manner.
 Display posts from previous: All Posts1 Day7 Days2 Weeks1 Month3 Months6 Months1 Year Oldest FirstNewest First
 All times are GMT Page 1 of 1

 Jump to: Select a forum The Antenna Forum----------------Questions Related to Antenna TheoryNews
You cannot post new topics in this forum
You cannot reply to topics in this forum
You cannot edit your posts in this forum
You cannot delete your posts in this forum
You cannot vote in polls in this forum